Manometers MP KIMO MP KIMO MP Data Sheet ( KB) For detailed information please refer to: [email protected] · Environnement SA. This is the MP Thermo-Anemo-Manometer. It is an easy to use meter which can take complex measurements. The MP is designed to measure Air. KIMO Instruments portable thermo-anemo-manometer with backlit digital display. Functions: pressure, temperature, air velocity and airflow with Pitot tube, CO.
|Published (Last):||19 December 2018|
|PDF File Size:||15.88 Mb|
|ePub File Size:||18.33 Mb|
|Price:||Free* [*Free Regsitration Required]|
Kimo MP200 Thermo-Anemo-Manometer
V — General informations Maintenance KIMO mo200 calibration, adjustment and maintenance of all your instruments to guarantee a constant level of quality of your measurements. If K factor is selected, you must enter value.
Battery When battery indicator flashes it is recommended to change the batteries: Preview of tables of points of datasets Dataset summary You can display tables of points of datasets performed on the device. For measurement launching, click on OK with the access key.
KIMO MP manometer buy in Kamenskoe
Alarm threshold corresponds to the right value on the bargraph. When the measurement is fixed, press Min Max to access to maximum and minimum values of the measurement. Configuration If you use thermocouple probes, you must enter type into the Configuration sub-function. Preview of tables of mo200 of datasets Dataset summary You can display tables of points of datasets performed on the device.
Kimo MP200 Multi Function Manometers
Comments to this Manuals Your Name. Wire probes with Smart-plus system Wire probes with Smart-plus system are automatically recognized when connected to the instrument. This datasets can’t be set using Datalogger Software. Enter dataset time and interval of time between 2 measurements by selecting Period with access key. Table of Contents Add to my manuals Add. Repeat this operation for the month, year, hour and minute. If the probe has been just connected, the screen indicates the remaining time of warm up then the measurement screen appears, indicates the kmio and places it on a bargraph.
This function can also be automatically performed by using the ,p200 valve function.
You can access this function selecting COmax with the access key CO max The CO is measured on an adjustable period, the maximum value measured in this period is called CO max. In regards of Quality insurance norms, we recommend that the instruments are checked once a year.
This menu allows to view probe Probe mimo information plugged to C2, Module, C1 or wireless connections. Remove the front part at the back of the instrument. In regards of Quality insurance norms, we recommend that the instruments are checked once a year.
You must enter duration: To print the manual completely, please, download it. The numbers of points is displayed.
Create or launch a continuous kmio Select dataset A continuous dataset can be carried out using MP and is composed of several dated measuring points. Don’t show me this message again. Click on OK to validate.
Summary screen of selected dataset is displayed. Table Of Contents actoolsupply. To modify the period, press Period with the access key. The ”Probe” menu only appears when probes or module are connected.
Summary screen of selected dataset is displayed From this screen, you can: This manual also for: From this screen, you can: