Dec 31, Operating and service manual for the HPA Synthesizer/Function Generator. Complete with schematics. Who I HEWLETT mllnM PACKARD OPERATING AND SERVICE MANUAL MODEL A SYNTHESIZER/FUNCTION GENERATOR Serial Numbers: AH. HP / Hewlett Packard / Agilent Used A Synthesizer/Function Generator, Refurbished – Used Test Equipment at Test Equipment Depot.
|Published (Last):||23 October 2010|
|PDF File Size:||6.62 Mb|
|ePub File Size:||16.90 Mb|
|Price:||Free* [*Free Regsitration Required]|
Table briefly summarizes the engineering changes that have brought A3 to its current revision. The output signal for each function is centered about zero volts unless a dc offset has been programmed.
Connect A signal output to the AC voltmeter via a 6 ft.
Maximum and minimum amplitudes are shown in Table Change attenuator to 12 dB. Enable Signals For Function A General miscellaneous parts are listed at the conclusion of Table Bits 3 through 0 in the status byte can be masked so that the corresponding conditions will not cause a service request.
HP 3325A Calibration Manual
Function and location of the part. Change A amplitude to 1 V p-p, and change oscilloscope vertical to. Fall time should be less than 20 nanoseconds.
Modulation depth at full output for each range: This also applies for Figure R A2C4 ? If the instrument fails any of these checks, it should mznual referred to qualified service personnel for repair. Dimension a string variable for each parameter you want to interrogate. Sine janual Square Wave: If you find any mistakes or if you know of a way to improve the procedure, please let us know.
Set the digital voltmeter to measure dc voltage.
Full text of “HP A Synthesizer/Function Generator Operating and Service Manual”
Close-In Spurious Signal Test. Change the A function to positive slope ramp. Delay Percent of Ramp. Connect the A output through a ohm feedthru termination -hp- 1IC to the oscilloscope input input dc coupledor set the A input switch to 50 ohms.
AH A5 board revisions are identical in design and component layout. Connect the A and the high-speed digital voltmeter through the divider as shown in Figure B.
Approved for public release; distribution is unlimited. If you find any mistakes or if you.
HP A User Manual | 23 pages | Also for: B
Connect the equipment as shown in Figure A. Using the Modify keys, adjust the A phase to return the counter reading to the value recorded in Step f. Digital voltmeter reading should be – V to Programming Amplitude Units Conversion. This flag can be used to determine when the A has finished processing. Note that the performance test steps and adjustments referred to in this service note may not correspond directly with the steps currently found in Sections IV and V.
Amplitude Accuracy and Flatness.
Component replacement and internal adjustments must be made by qualified maintenance personnel. Returns A from remote to front panel control unless Local Lockout has been programmed.
HP 3325A Synthesizer/Function Generator Operating and Service Manual
Enter the program into the calculator. Integrator and Phase Modulation Circuitry It is recommended that no other equipment be connected to this HP-IB during this test.
General Interface Management Lines. A6 – Past to Present.