Dec 31, Operating and service manual for the HPA Synthesizer/Function Generator. Complete with schematics. Who I HEWLETT mllnM PACKARD OPERATING AND SERVICE MANUAL MODEL A SYNTHESIZER/FUNCTION GENERATOR Serial Numbers: AH. HP / Hewlett Packard / Agilent Used A Synthesizer/Function Generator, Refurbished – Used Test Equipment at Test Equipment Depot.
|Published (Last):||21 November 2012|
|PDF File Size:||3.58 Mb|
|ePub File Size:||5.34 Mb|
|Price:||Free* [*Free Regsitration Required]|
Search mmanual history of over billion web pages on the Internet. Earlier versions of the A, however, may differ in design from the instruments this revision documents directly. The delta symbols refer servicing personnel to the backdating section Section VII where specific information regarding the change is presented.
Box 69, Marysville, Washington Printed: Failure to comply with these precautions or with specific warnings elsewhere in this manual violates safety standards of design, manufacture, and intended use of the instrument Hewlett-Packard Company assumes no liability for the customer’s failure ho comply with these requirements. This is a Safety Class 1 instrument. The instrument is equipped with a three-conductor ac power cable. The power cable must either be plugged into an approved three-contact electrical outlet or used with a three-contact to two-contact adapter with the grounding wire green firmly connected to an electrical ground safety ground at the power outlet.
Providing the finest test equipment solutions since 1992
The power jack and mating plug of the power cable meet International Electrotechnical Commission IEC safety standards. Operation of any electrical hhp in such an environment constitutes a definite safety hazard. Component replacement and internal adjustments must be made by qualified maintenance personnel.
Do not replace components with power cable connected. Under certain conditions, dangerous voltages may exist even with the power cable removed. To avoid injuries, always disconnect power and discharge circuits before touching them.
HP A User Manual | 23 pages | Also for: B
Instructions contained in the warnings must be followed. Indicates dangerous voltage terminals fed from the interior by voltage exceeding volts must be so marked. For protection against electrical shock in case of a manuaal. Low-noise or noiseless, clean ground earth terminal. Used for a signal common, as well as providing protection against electrical shock in case of a fault.
A terminal marked with this symbol must be connected to ground in the manner described in the installation operating manual, and before operating the equipment.
Frame or chassis terminal. Alternating current power line. Direct current power line. Alternating or direct current power line.
It calls attention to a pro- WARNING I cedure, practice, condition or the like, which, if manula correctly per- formed or adhered to, could result in injury or death to personnel. It calls attention to an operating procedure, practice, condition or the like, which,if not correctly performed or adhered to, could result in damage to or destruction of part or all of the product.
The NOTE sign denotes important information. It calls attention to procedure, practice, condition or the like, which is essential to highlight.
HP 3325A Calibration Manual
Instrument and Manual Identification Frequency Display and Resolution. Auxiliary Output Sine Function Only High Voltage Output Option Sweep X Drive Output.
Sweep Z Blank 33325a. Basic Device Communication Capability. Universal and Addressed Commands Placing the A in Remote. Programming Data Transfer Mode Programming Binary On or Off Functions Programming Execution Functions Programming Amplitude Units Conversion.
Using manuxl Interrogate Capability. Meta Messages Block Diagrammed. B-l Section Page IV. Triangle and Ramp Verification.
Full text of “HP A Synthesizer/Function Generator Operating and Service Manual”
Output Level and Attenuator Check. Close-In Spurious Signal Test. Integrated Phase Noise Test. Amplitude Modulation Envelope Distortion Test —.
Square Wave Rise Time and Abberations. Option High Stability Frequency Reference. Sine Wave Amplitude Calibration. Change Sheets and Service Notes Service Group C – Control Circuits Control Circuits Service Group C. Service Group D – Voltage Controlled Mixer Service Group H.
Service Group F – Fractional N Attenuator Service Group L High Voltage Output Option A4. Service Group H – Mixer Power Supplies Service Group O Sine Amplitude Control Path. Service Group K – Output Amplifier Service Group L – Attenuator Power Line Voltage Selection Service Group O – Power Supplies Adjustments Required After Repair.
Logic Troubleshooting by Signature Amplitude Limits of AC Function. High Voltage Output Amplitudes Maximum DC Offset General Interface Management Lines Definition of Meta Messages. A2l Al Board Revisions.
A21 A1 Board Revisions. Summary of A Programming A14 A4 Board Revisions. Test Equipment Required for A23 A7 Board Revisions. Test Equipment Required For Attenuation and Voltage Ranges. Test Equipment for Troubleshooting. Front and Rear Panel Controls . Processor Interrupt Circuitry Serial Schematic and Board Location of R11 and Harmonic Distortion Verification Integrated Phase Noise Test Amplitude Accuracy and Flatness. Triangle and Ramp Linearity Test. Integrator and Phase Modulation Circuitry X Drive Linearity Test.
Simplified Illustration of Ramp A and Below. Sine Amplitude Control and Amplitude X Drive Ramp Output. Mixer Driver Circuitry Serial Numbers Amplitude Control Circuitry Serial Signature Analysis Test 4. Keyboard and Display, A